High-definition area scan (99 x 99 points) on a well of a 96-well microplate
October 03, 2011Download
Related Products: Gen5 Funktionen
Gen5 version required: 2.0 or higher
One of the problems when reading cell assays in microplates is that the cell population is often not homogeneous at the bottom of the wells and measuring only at the center of the well leads to inaccuracies. The area scanning read mode allows correcting for this artifact by providing an averaged result over the entire area of the well. This sample file demonstrates Gen5’s high definition (99 x 99 points) area scanning feature on one well of a 96 well microplate. This data allows seeing where the cell layer has been disturbed by the liquid handling process.
99x99 area scan of a well of a 96-well plate. Signal drop at the bottom was confirmed under the microscope as a damaged area in the cell layer.